Andreas Kaufer, Florian Kerber, R Hanuschik, F Patat, M Peron, M Romaniello, M Sterzik
[en] Thinned CCD detectors display fringing which arises from the interference of multiply reflected light in the layers of the CCD. If the layer construction - the thicknesses and refractive indexes of the layers - is known, then the observed fringing can be accurately modelled and used to correct imaging and spectroscopic data for its effects. In practice the specifications on the actual deposited layer thicknesses may not be known to sufficient accuracy to predict the fringe behaviour. Thus calibration data, in the form of monochromatic flat fields, is required and can be modelled using the technique outlined by Malamuth et al. 2003, which has been applied to ACS CCDs.